Question about Zeiss Microscope
Since there's only a very small amount of light reaching the lens system of microscopes when on high powers, A special technique (called >>>oil immersion
Not using this technique must be the cause of the problem you are experiencing.
eHow will guide you through the exact procedure of using the oil immersion technique: http://www.ehow.com/how_8431358_focus-high-power-objective-microscope.html
Good luck in exploring the micro world!
Posted on Jul 09, 2011
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Jul 03, 2011 | Computers & Internet
Dec 01, 2010 | National Optical 131 Monocular Microscope
Sep 17, 2010 | Zhumell LOMO P-111 modular brightfield...
Jul 25, 2010 | Premiere Microscopes MRJ-03 Microscope
Jan 04, 2010 | Science & Laboratory
Mar 05, 2009 | Science & Laboratory
Nov 25, 2008 | Nikon 7314 Binocular Microscope
Dec 30, 2007 | Tasco 750 Monocular Microscope
321 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: